The 1/8 DIN unit is a controller with an extensive range of options. Either PID, ON/OFF or Motorized valve control can be configured, satisfying both. Contents. Issue Nov Applies to and controller software version PDS and INSTANT ACCURACY are trademarks of Eurotherm.” Issue Eurotherm by Schneider Electric Temperature Controller Programmer. See price below. 1/8 DIN panel size PID Temperature Controller. Specifications.
|Published (Last):||22 September 2014|
|PDF File Size:||12.39 Mb|
|ePub File Size:||6.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Eurotherm 2408 Temperature Process Controller
Eurohherm my guess is that it eurotehrm stops at C because it is the limit of the power supply. If I skip the error and let it try to read from the device, then I get timeout errors. Rolf Kalbermatter Averna BV. Like I said earlier, eurotherm 2408 works in iTools so I’m confident it’s wired properly. Alternatively email eurotherm 2408 to support eurotherm.
The Labview vis uses Visa. I cannot set the SP manually.
Make the above changes and try Autotune again to get the best PID values for your load. Can you change the device to use uerotherm local setpoint instead the Remote setpoint?
The Eurotherm controllers can be eurotherm 2408 with either RS or RS interface it’s a module that can be eurotherm 2408 inside the controller. Cannot get labview eurotherm 2408 iTools to communicate with a controller. I also spent quite a bit of time for setting up Eurotherm RS Comm. So please make sure you eurotherk installed serial Visa as per the ET24xx driver install instructions. I just tried to run in Local setpoint mode with the same result. Hi Dimitri By restarting the computer and opening iTools alone Eurotherk managed to get contact to the device so that is great, thanks a lot!
Eurotherm 2408 have the same one.
Make a note of the iTools Eurotherm 2408 address of the Setpoint parameter. The set point example does not report any errors but it just does nothing.
Communication with Eurotherm
Best Jakob How does this make you feel? Message 5 of 6. If you used Labview to connect to the device using the same serial port then iTools will not detect the device. Eurotherm 2408 error is generated in the initialize routine and the message is displayed below. Applications Engineering National Instruments.
I cannot open the eurotherm 2408 as have 248 8. I hope this helps.
Eurotherm 2404/2408 Controller/Programmer
I downloaded the et2xx labview drivers and I cannot get either of eurotherm 2408 two example programs to work in labview. Oh, and I am trying to control the temperature of a sample in UHV. Hi Jakob, During Autotune the PV will go few degrees above the setpoint as part of the tuning process. I will explain details if I get a eurotherm 2408.
Hi Dimitri It is working now! Sounds like we have quite a few different things going on here, so lets backtrack a bit and see where were getting the communication issues. Message 1 of 6. Another point to eurotherm 2408 is that if you are using Serial communications then you cannot use iTools and Labview at the same time as the first application that connects to the will eurotherm 2408 the Eueotherm port.